Plate Material Evaluating Cell
This quite handy cell is developed in order to
evaluate plate material such as metal,
semi-conducting plate, etc. In evaluation, a sample
plate is sandwiched between cell blocks.
|011951||Plate Material Evaluating Cell||1|
|Teflon Cell [Body]||1|
|Teflon Cell [Base]||1|
|Screw 20 mm||2|
|Purging tube 1 m||1|
|002222||Pt counter electrode for VC-2 cell||1|
[Optional Products] Reference Electrode
Users can select a reference electrode separately for this cell from :
|012167||RE-1B Silver-Silver chloride reference electrode|
|012171||RE-7 Non Aqueous reference electrode|
|002058||RE-1C Silver/Silver Chloride Reference Electrode (Saturated KCl)|
|002056||RE-2B Calomel Reference Electrode|
|002057||RE-2C Reference Electrode|
|011503||RE-6H Reference Electrode Kit available in alkaline solution|
How to Set Up
1. Place a plate sample on a lower side teflon
block of the cell.
Applicable plate dimensions : W: 10-20 mm, T: ~3 mm
2. Put another teflon block on the sample, and drive two screws evenly to fix the plate. Liquid contact area depends on inner diameter of the O-ring. Deliver approximately 1 ml of electrolyte after fixing the plate.
3. Attach a Pt counter electrode and a reference electrode to the cell by setting them onto a cell cap. We would recommend you to twist up the Pt part appropriately if the counter electrode is longer.
<<Home ALS Main Page